LABVIEW EUROTHERM DRIVER DOWNLOAD
Message 4 of 6. This was not possible to get working although I didn’t have a chance to go very deep into debugging with this. With this configuration, I made it work. The REM should flash when there is a connection however I cant get it flash so I assume the problem is in connection. Now I have a strong suspicion that the Eurotherm driver you downloaded was actually developed and tested with a controller with RS interface.
|Date Added:||28 August 2007|
|File Size:||67.59 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Message 1 of 6. With this configuration, I made it work. Or should I buy the serial protect from NI? Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.
Labview Driver now Available for Eurotherm nanodac – New England Temperature Solutions
Message 3 of 6. We primarily host it for convenience. Unless your interface card eurothsrm these resistors built in, it is likely the problem why you can’t get this work. Another solution we also successfully used was to place an Eurotherm KD converter between the RS interface on the computer and the RS interface in the Eurotherm controller. Instructions for that are here: The REM should flash when there is a connection however I cant get it flash so I assume the problem is in connection. Also, just wonna make sure that whether the serial port I am using is compatible.
Also, the et24xx driver maybe shouldn’t be assumed to work.
Sounds eurotherrm we have quite a few different things going on here, so lets backtrack a bit and see where were getting the communication issues. Most Active Software Boards: The Eurotherm controllers can be delivered with either RS or RS interface it’s a module that can be replaced inside the controller. Message 2 of 6. I found that terminal registor cause significant voltage drop, so I do not use any termial registor.
Now I have a strong suspicion that the Eurotherm driver you downloaded was actually developed and tested with a controller with RS interface. However time and other constraints durotherm not allow us to debug that deep enough to make it work as a full RS connection. On software configuration, here is the list.
Message 6 of 6.
Eurotherm nanodac Controller – Ethernet Driver for LabVIEW – National Instruments
I hope this information will help you make your system work. Is it really necessary for me to use the serial port from NI instead of using random brand serial port?
Also, I am thinking if the issue is due to the incorrect serial port I am using. What did work is to replace the RS interface in the controller with a RS module and connect it to an RS port on the compactFieldpoint.
The unit I got from the client to do some testing in our office appeared to be defect, and testing in the factory installation is a hassle. This was not possible to get working although I didn’t have a chance to go very deep into debugging with this. Some of the NI plugin boards do have programmable bias resistors that can even be switched on and off through a VISA attribute node if I’m not mistaken.
Message 5 of 6. Message 4 of 6.
Rolf Kalbermatter Averna BV. Applications Engineering National Instruments. Thanks for the reply The loopback works well and the LabVIEW driver I am using is directly downloaded from NI website et24xx so it can be assumed that the problem is not on the code?
Eurotherm et24xx Controller
I encounter a problem with my project. Can anyone give me a hand? I also spent quite a bit of time for setting up Eurotherm RS Comm.